# data[].attributes.body
Atividade Prática de Medição Simulador
Atividade com Instrumento!

# data[].attributes.headline
                                                     Erros de Leitura!

# data[].attributes.subject
                                                     Erros de Leitura!

# data[].attributes.attachment_link.readable_text
Virtual dial caliper in millimeter - simulator of use

Use, reading and interpretation of dial caliper in millimeter

0.01 mm resolution (one hundredth of millimeter)

Caliper is a linear measuring instrument with a scale and a slider that slides on it. It is designed to take external linear measurements by contact. It can also measure internal dimensions, depths, bumps, among others.

The main scale of this simulator is graduated in centimeter, which is divided into tenths (millimeter).

In the dial, the pointer gives a full turn (360 °) every millimeter of advance and its dial has a hundred divisions. In this way, each stroke that the pointer "sweeps" represents the feed rate of 1/100 of the millimeter or one hundredth of a millimeter.
One advantage of adopting the watch to increase the resolution of the caliper is that it allows the display to rotate, after unlocking, to adjust the instrument and facilitate the inspection of the tolerance of several pieces without the need to interpret each one.

This instrument, as a rule, can not replace the micrometer with hundredth resolution, since its accuracy

(imprecision between presented and actual value) is greater than that of resolution.

Caliper with virtual dial in millimeter with 0.01 mm resolution - simulator of use and practice of reading and interpretation

The monitor resolution doesn't allow that the virtual instrument interpolate every possibility of measure, if you wish to investigate these possibilities, interact with the simulator of dial indicator and transfer what you have learned.

# data[].attributes.attachment_link.title
Virtual Dial Caliper - One Hundredth of Millimeter - Simulator | Prof. Eduardo J. Stefanelli

# data[].attributes.attachment_link.description
Virtual dial caliper in millimeter - simulator of use Use, reading and interpretation of dial caliper in millimeter 0.01 mm resolution (one hundredth of millimeter) Caliper is a linear measuring instrument with a scale and a slider that slides on it. It is designed to...

# data[].attributes.attachment_link.readable_text
Metric vernier caliper, (calliper or pachymeter) in millimeter and vernier scale resolution 0.02 mm - Reading and Interpretation Practice Simulator

Metric vernier caliper in millimeter, 0.02 mm

Caliper is a linear measuring instrument with a scale and a cursor that slides on it. It is designed to take external linear measurements by contact. It can also measure internal dimensions, depths, bumps, among others.

The main scale of this simulator is graduated in centimeter, that is divided in tenths (millimeter). The vernier divide the millimeter in fifty (1/50), marking the 0.02mm (two hundredths of a millimeter).

See the main components of a vernier caliper clicking here .

To comprehend the science of measurement with caliper, we suggest that you read and interact with the pages:

* Simulator of reading and interpretation vernier scale in millimeter, tenth resolution

* Virtual vernier scale - simulator of use and reading, resolution two hundredths of millimeter 0.02 mm

Vernier caliper in millimeter with resolution 0.02 mm millimeter (two hundredhs of millimeter) simulator and practice of reading and interpretation
tips
Evaluate your progress:
Self-assessment: caliper in millimeter with 0.02 millimeter (hundredth)
Solve:
list of exercises of reading and interpretation of caliper with vernier in millimeter

Know other technologies of caliper in millimeter

# data[].attributes.attachment_link.title
Virtual Vernier Caliper - simulator in 0.02 Millimeter | Prof. Eduardo J. Stefanelli

# data[].attributes.attachment_link.description
Metric vernier caliper, (calliper or pachymeter) in millimeter and vernier scale resolution 0.02 mm - Reading and Interpretation Practice Simulator Metric vernier caliper in millimeter, 0.02 mm Caliper is a linear measuring instrument with a scale and a cursor that slides on it. It is...

# data[].attributes.attachment_link.readable_text
Virtual vernier caliper in millimeter - simulator of reading, interpretation and practice - vernier scale resolution 0.05 mm

Metric vernier caliper, (calliper or pachymeter), read in millimeter and vernier scale 0.05 mm

The vernier caliper, is a measuring instrument equipped with a scale and a cursor that slides in itself. In the cursor is recorded a second scale called vernier scale or nonius.

At one end of the scale there is a beak with a perpendicular face on it. On the cursor there is also a beak with a perpendicular face to the scale, and when those faces are in contact the trait 0 (zero) of cursor aligns with the mark 0 (zero) of the scale.

Calipers were designed to take external linear dimensions by contact. With less precision also measures internal dimensions, depths and protusions. In fact, for this reason it is also known as four-dimensional caliper.

To take external measures we position the object to be measured in contact with the face of the backrest of the fixed beak, the closest possible to the scale, without touching it, and gently slid the cursor with your thumb until the face of the movable backrest beak touch the object without hitting it, and without exerting too much pressure. Reading the whole value in mm is taken by comparing the mark position 0 (zero) of the cursor with fixed scale, and the decimal part by observing which mark of the cursor aligns with a trait of fixed scale. See page Vernier scale simulator to better understand this method.

This virtual vernier caliper is the replica of the metrology tool used in the metalworking industry.

The main scale of ths simulator is graduated in centimeters (1cm), which is divided by ten (millimeter). The vernier scale divide the millimeter by twenty (1/20), measuring 0.05 mm (five hundredths of a millimeter).

Virtual vernier caliper - how to use the vernier caliper measurement tool - vernier scale resolution 0.05 mm - simulator and practice of reading and understanding

To understand the science of measurement with vernier caliper, we suggest that you read and interact with the pages:

* Reading and understanding simulator of vernier scale in millimeter, decimal resolution

* Virtual vernier scale - simulator of use and reading, resolution five hundredth of millimeter 0.05 mm

evaluate your progress:
Self-assessment: caliper in millimeter with 0.05 mm resolution (five hundredths)
Exercises
list of exercises of reading and interpreting of vernier scale / vernier caliper in millimeter

learn about other technologies calipers in millimeter

# data[].attributes.attachment_link.title
Virtual Vernier Caliper - simulator in 0.05 Millimeter | Prof. Eduardo J. Stefanelli

# data[].attributes.attachment_link.description
Virtual vernier caliper in millimeter - simulator of reading, interpretation and practice - vernier scale resolution 0.05 mm Metric vernier caliper, (calliper or pachymeter), read in millimeter and vernier scale 0.05 mm The vernier caliper, is a measuring instrument equipped with a scale and a...

# data[].attributes.body
1. Medidas Lineares;

2. Medidas Internas;

3. Medidas Externas e;

4. Medidas de Profundidade.

# data[].attributes.headline
Metrologia Para Mecânica - Utilizando Instrumento Paquímetro!

# data[].attributes.subject
Metrologia Para Mecânica - Utilizando Instrumento Paquímetro!